Spectrum Adulterant Screen is designed to enhance FT-IR/NIR Spectrum 10 Software, allowing the development and execution of adulterant screening methods for suspected materials of concern. The algorithm takes into account sample variability and provides a simple method which adapts to new threats. Requires Spectrum v10.4.2 or later. Standard and ES compatible.
Please enter valid quantity
Please log in to add favorites.
NULL OR EMPTY CART
Increasing demand and high-market value of ingredients and food has led to numerous adulteration and fraudulent schemes which undermine the quality attributes and safety of food. Adulterant Screen offers simple method development with a powerful algorithm to ensure detection, identification, and estimation of concentration of materials which have been fraudulently or accidentally introduced into a product.
Adulterant Screen is the ideal solution for laboratories needing to test products to verify that they are the stated product, and that they are free from contamination or adulteration. There is no need for extensive software training or methods development. It is particularly useful for, but not limited to, the screening of:
The Spectrum 10 software is used to collect IR or NIR spectra representing “good samples” for a product. This consists of multiple spectra representing many sources of variation of the product, such as different batches, sources, grades of material, etc. These spectra define the boundaries of the acceptable product. Spectra of all known adulterants as pure materials are then collected, which define the Adulterant Spectra.
Accelerated Method Optimization
Adulterant Screen methods can be rapidly developed to detect and semi-quantitatively predict the level of adulterants. Quantitative predictions are based on the pure spectra of adulterants and, do not require the preparation of calibration standards and methods. A suite of pre-processing algorithms are included to assist with common sampling variability issues and improve method robustness, accounting for spectral baseline effects and spectral regions of poor signal to noise. Blank regions can be incorporated to exclude spectral regions that do not contribute useful information to the method.
Analysis is future-proofed, since new adulterants can be simply added by scanning a single spectrum of the pure adulterant and adding it to your existing list of adulterants. No lengthy recalibration is required. Adulterant Screen can be used on demand as a data processing command within Spectrum 10 software. It can also be easily incorporated into Spectrum 10 macros and Spectrum Touch methods to provide simple turnkey analyser operation with PerkinElmer IR and NIR systems. These methods can incorporate additional Spectrum Quant methods to quantify materials in the product if it passes the Adulterant Screen.
|Product Brand Name||Spectrum Adulterant Screen|
|Part Number||Product Description||Users||STD/ES|
|L1101026||Spectrum Adulterant Screen 1-user license||1||STD/ES compatible|
|L1101034||Spectrum Adulterant Screen 5-user license||5||STD/ES compatible|
|L1101035||Spectrum Adulterant Screen 10-user license||10||STD/ES compatible|
Application Note, FT-NIR, Infrared, Near-Infrared, FT-IR, Spice, Spice Adulteration, Adulterant Screen, SIMCA, Verify, Spectrum Touch, Frontier
Infrared, FT-NIR Spectrometry, Near-Infrared, NIR, yogurt, yoghurt, melamine adulteration, Fronteir, Adulterant Screen
Application Brief, Single NIR Measurement for the Detection of Adulteration and Measurement of Important Parameters in Cocoa Powders, Cocoa Powder, Near-Infrared Spectroscopy, NIR, Adulterant Screen, Spectrum Touch
Infrared, Near-Infrared Spectroscopy, IR-NIR, Adulterant Screen, Milk, Nitrogen urea detection, Frontier NIR, NIRA II sampling accessory
Adulterant Screen, Spectrum Software, Spectrum 10 Software, IR Spectroscopy, FT-IR Spectroscopy, NIR Spectroscopy, Adulterant Spectra, Materials Spectra, Spectrum Adulterant Screen, Spectrum Adulterant Screen ES, Spectrum 10 Enhanced Security