Please enter valid quantity
Please log in to add favorites.
NULL OR EMPTY CART
As the blue septa used at lower operating temperatures the non-stick coating is not required to prevent inlet adhesion. They offer a low level of inertness and for routine GC applications providing optimum performance at lower operating temperatures.
|Model Compatible With||Clarus 680, Clarus 480, Clarus 590, Clarus 580, Clarus 690|
|Outer Diameter||11 mm|
|Product Brand Name||Blue Injection Port Septa|
|Product Group||Injection Port Septa|
|Quantity in a Package Amount||50.0|
This technical note outlines the inlet septum replacement procedure for the CAP injector assembly (Item #: N6100007, 120V; N6100137, 230V) for the Clarus® 500/580, & 600/680 GC platforms. The injector assembly is designed for quick access to the inlet septum for routine maintenance and easy integration into daily workflows. Figure 1 provides a diagram of the injector assembly as viewed from the top of the instrument chassis, highlighting placement of the septum cap (Item #: N6100153) containing the septum within.
When compared to the exacting attention that gas chromatographers pay the analytical column, it often seems that the inlet septum receives relatively little consideration. In fact, many chromatographers are surprised to discover that GC instrument manufacturers recommend daily inspection and/or replacement of the inlet septum to maximize instrument performance. Although inexpensive compared to other chromatography consumables, a freshly replaced septum is essential for maintaining system isolation from the outside environment.
A poorly maintained inlet will cause the instrument to suffer from diminished sensitivity, shifting response factors, asymmetrical peak shapes, and an overall poorer precision and accuracy compared to a well-maintained one. When older, damaged septa are cored by the syringe needle these cores of septa can drop into the inlet liner, blocking the flow of gas and releasing bleed volatiles onto the column. Such detriments impact system sensitivity and inevitably result in unscheduled maintenance and more frequent calibrations.
This technical note outlines the inlet septum replacement procedure for the capillary split/splitless (CAP) injector (Item #: N6120146) for the Clarus® 590/690 GC platform.