This product is not available for online purchase.
The TotalChrom™ Chromatography Data Systems (CDS) manage the growing volume of chromatography data quickly and efficiently. TotalChrom data systems have the industry’s best expandable architecture and can be configured in stand-alone, thin-client, or client/server environments.
For regulated environments, TotalChrom enables compliance with 21 CFR Part 11 regulations with ease. Even at the highest throughput levels, TotalChrom handles peak detection, integration and analysis with sophisticated algorithms, refined through years of experience in chromatographic data processing.
The software features TC Publisher™ which is simply the world’s best chromatography reporting package available. With its flexible reporting tools, you can automatically generate an endless variety of reports.
TotalChrom is backed by PerkinElmer’s world-class service and support organization to ensure virtually flawless installation and training. A wide array of system IQ/OQ/EOQ services and system support/maintenance contracts are available to support your validation needs.
TotalChrom has won the Scientific Computing and Instrumentation Reader’s Choice award for nine consecutive years.
What would you like to do?
Review Product Details
TotalChrom C/S delivers the power to handle challenging
chromatography, networked data acquisition, monitor real-time data
acquisition from anywhere, generate and distribute custom reports,
and quickly and easily build complete methods and sequences.
Additionally, when configured with a PerkinElmer Clarus® 500
& 600 GC, Autosystem (XL) GC or a PerkinElmer Series 200 HPLC,
the TotalChrom software serves as both a controller and data
manager for the overall system. Compatible with all commercial LCs
and GCs, TotalChrom is the best choice for chromatography data
handling in demanding multi-user, multi-instrument laboratory
PerkinElmer recognizes the importance of proper training to ensure that our customers benefit fully from the capabilities of our instruments.Our Training Academy offers many training options; such as web based training, classroom or customized courses, to help you get the most out of your time in the lab.
Topics range from theory to techniques relating to specific methodology, optimization, and familiarization with instrumentation and related software. Courses are offered at various locations throughout the Americas, as well as Seer Green, England and Turku, Finland.
Learn more about training.
How do I customize the default method for all users
Default method templates originate from the System folder in Penexe\TCCS\VER6.1.2\Config\Sys. When a new user is created a folder is generated in Penexe\TCCS\VER6.1.2\Config\User, with that users logon name. The template method from the SYS folder is copied into this users folder. That method will be used as the default method whenever the "New" method option is selected. If specific values are preferred for default methods, the SYS folder method can be edited with the Method Editor and saved. As users are created this template will be copied out to all users personal folders. If users already exist then the template method can be used to overwrite the existing default.mth file present in each users folder. The original default.mth file should be save as a backup in a safe location. Instruments should not be assigned to default methods. It may be preferable to set the file to Read Only to prevent accidental modification.
How do I prevent broadcast error messages to all of the TCCS clients when I remove power from a controlled instrument?
You must first release control of the instrument and then detach it. These commands can be executed from the Run button on the Navigator screen or Run on the menu bar in Navigator.
I am using Turbochrom Client/Server. My Win95 client will not let me create a sequence because I can not see any grids. The WinNT client works just fine. What might be happening here?
There is a probable mismatch in some of the older DLL files that are referenced in the Turbochrom Client/Server grid control modules. You would need to update these and re-register them. This is done outside of the Win95 environment (by rebooting the workstation in DOS only mode). The following is the procedure for doing this.On some Win95 machines the Microsoft dll, OLEAUT32.dll is incompatible with the Turbochrom Client Server grid control. The symptom is no grid showing up in the sequence editor. At release time there where issues around installing this automatically during setup. With the current release code if you have this problem follow instructions below. The file OLEAUT32.dll is currently on the TC C/S CD. The following applications use the OCX grid control:
All GUI applications use OcxGrid.dll as their source for the grid control API. The following files are required under c:\winnt\system32 on the clients target machine. TO DATE THE CRITICAL FILE IS OLEAUT32.DLL. IF THE GRID IS NOT FUNCTIONAL AFTER INSTALLATION COPY THIS FILE FROM THE INSTALLATION CD INTO THE WINNT\SYSTEM32 OR WINDOWS\SYSTEM DIRECTORY (if Win95). THEN FOLLOW THE COMMAND BELOW FOR MANUALLY REGISTERING THE CONTROL.
|Sep 13 ||9:47 ||1013520 ||MFC42.DLL ||4.2.6256|
|Aug 2 1996|| ||267536 ||MSVCRT.DLL ||4.20.6201|
|Aug 8 1996|| ||676624 ||ole32.dll ||4.00|
|Aug 8 1996|| ||473872 ||oleaut32.dll ||2.20.4049|
|Jun 19 1996|| ||76048 ||OLEPRO32.DLL ||4.2.6068|
|Oct 31 ||10:57 ||16896 ||stdole2.tlb ||2.20.4054|
|Feb 4 ||16:03 ||790528 ||VCF132.OCX ||126.96.36.199|
If the installation for Turbochrom is locked from updating the registry, try to update the registry for the grid control by hand. In this directory, there is a windows application (regsvr32.exe) that updates the registry for ActiveX controls.
For NT, the syntax for this command is: \penexe\tccs\ver6.0.0\bin\regsvr32.exe c:\winnt\system32\vcf132.ocx
For Win95, the syntax for this command is: \penexe\tccs\ver6.0.0\bin\regsvr32.exe c:\windows\system\vcf132.ocx
Of course, substitute specific directories that apply to your system. A window will be displayed indicating success or failure of this command. If it fails for any reason, make sure the .dll listed above are in your path correctly.
How can I install the Turbochrom Client Server software on a workstation without a CD Drive?
If the workstation doesn’t have a CD, you have two ways to install the TCCS software. Copy the entire contents of the CD to a directory on another machine and share that directory. Copy the CD so as to preserve the directory structure. Next, on the machine you want to install the software, map that directory (using Explorer). For example.. On machine A create a directory , say TCFILES. Copy the CD contents to that directory. On the machine you want to install the TCCS software, map that directory…. N: to point to the directory on that machine. Then simply run the setup.exe on the mapped drive. Alternately...On a machine (such as the server), share the CDROM drive as above, and map that drive on the machine you wish to install the TCCS software.
To view our Frequently Asked Questions